|7032||Red HK not coming up (timeout talking to dataprobe)|| the iBootBar dataprobes have sometimes required someone to log in and type
One thing to do is to make sure we are cleaning up the socket to the device everywhere, even on errors.
|6800||OB hung with R (8.0s guiding failed) and B (GOYA rotator warning)|| no issue with the rotator, this problem is red controller had a hardware failure but somehow missed the STOP command that the software issued to recover. The observer issued a STOP command, but didn't wait until it was done to issue another STOP.
The hang should be fixed in version 2.15, released Sep-2018.
|6756||LBC pause/resume behavior|| UI reported number of exposures taken as greater than total requested
why is this only showing up when using one channel?
|5715||stars trailed on first LBC image after a dither|| John says this is likely due to LBC not checking for collimation update to finish; this is a one-sided LBC problem
Should be fixed in version 2.15, released Sep-2018.
|5302||Suggested changes/reorg of LBC image headers||enhancement|
|5205||OB progress is incorrect after pause/resume||bug|
|Timeout error on motion of filter wheel||We see this with wheel #2 for both red and blue. Seems like mostly it's a move command that doesn't get received on the motor because we timeout and then check and it hasn't moved at all.|
|4705||Add pointing offsets to LBC headers|| enhancement
Waiting on Olga for specifics, then CB can add to the TCS public interface.
|4642||Add notification on failures|| enhancement
pop up on tracking stopped - no, we do NOT want to stop operations
added ERROR message when tracking background is < 1 should it be identically zero? - cannot do a pop-up without stopping the OB. Fixed so it waits til it has tried up to the max exposure time before it complains
Added Z4 to GUI.
Make the background of the ? red -- needs testing.
Olga also added the "shutter seems to be not working" to this IT - changed this to return an ERROR - so it does not go to
how do we test the science image? don't have to, they don't have this problem
|4566||Red Rotator communication problem|| bug : When the rotator failed in a preset, it tried to retry indefinitely - it re-tried ~680 times before they stopped it. It is not really obvious to the operator that is happening, because it automatically retries and claims
|4236||hung at unitialization of blue HK||this is the HK timeouts - cannot seem to fix this within the software, but a lbckill/lbcstart fixes it!|
|3833||audible alarm on errors||enhancement|
|3816||Scale R/B exposures separately||enhancement|
|3759||Blue tracker images suddenly all zeroes|| bug
Happens on both sides. Goes away with an OFF/ON toggle. Sometimes it's not zero, but just funky.
Also look at adding a "reset" that power-cycles the electronics, but not the windows PCs.
|3595||Rotator position on first dither||still seeing problems as of Jan-2012|
|2765||tick marks are sticky on the UI|| I did some testing - this is all controlled by the browser settings and cookies and what you submit to the GUI. If you check mark, but do not "submit" by pressing PLAY or TURN ON or TURN OFF, then the settings will reset to the last time that happened in this browser environment. If you clear and restart, nothing is sticky - if you kill and restart, it remembers.
Olga notes in the IT that running dofpia and the check marks erasing? DOFPIA "submits" the http form so the checkmarks will fill in based on that.
|2631||too few downloaded pixels error||This is not mine, but I am monitoring the problem.|
|2573||LBCB/LBCR First Image of a Sequence has small Rotator Offset (~0.5 deg.)||reopened in Sep-2013; AdP says maybe to track Parallactic angle to open shutter delay.|
|2546||WCS info in LBC tech chip headers|| enhancement
Waiting on Olga to provide
|2397||RPC Client/Host Comm Error|| bugs
multiple problems documented here - guiding lapses, missing images, inhibit counter on restart, etc.
This became a big issue coming out of summer shutdown 2013. Added better logging, retry during preset.
Need to: release