On site: David, Barry
Tucson: Jenny, Olga, Shane, Andrew, John H., Michelle DLP, Doug, Kellee.
Local time (MST) used throughout.
20:10 David is correcting pointing and collimating.
20:23 Going to telluric for Barry\x92s first target.
20:28 12 deg twilight.
20:29 0\x94.9 - 1\x94 on guider.
20:31 DX LUCI WFS looks a little odd. John suspects a bad flat.
20:36 Using LS_2.0_1.5, centering in 1\x94.5 slit. (Lower slit as displayed.) Centered target on 1009.24, 517.01. Offset required was +7\x94.51, -59\x94.45.
20:40 Starting telluric observation.
20:43 Going to science target.
20:45 Starting acquisition.
20:48 LUCI2 error \x91no mask used\x92...
20:52 Having double checked the script we are calling Dave T.
21:05 Michelle leaves for the night.
21:10 Making some very small rotator changes and LUCI2 detector reads while Dave works. (To test the acquisition rotation sign issue.)
21:18 Looks like LUCI2 is out of action for the moment. Switching to LBCs.
21:50 moved to LBC first target: Bootes1, transiitting
22:52 Ran test of wrs but got error.
% WRITE_JPEG: Error opening file. Unit: 102
File: save/FINDPUP_results_2016Jun 7_04:50:15.jpg
No such file or directory
23:00 Running dofpia on Bootes1
23:10 dofpia hung, the data files never appeared in newdata.
23:16 Running dofpia again. Z22 runaway
23:37 Running dofpia again. Z22 on LBCB continues to runaway. Quit it and allowed observations to proceed
23:47 Bootes 2 started, LBCB ~ 1.3\x94, LBCR ~0.8\x94
22:35 Struggling with dofpia. It thinks the seeing is too bad on the blue side to correct the spherical, but red has worked fine.
22:44 Running the co-pointing script.
22:53 Going to Bootes 1.
23:09 Dimm is ~1\x94, LBCB images showing around 1.4\x94 in USpec , but LBCR ~1.0\x94 in RBessel
23:18 Running dofpia again
00:16 Switching to Christians targets. Large gradient on SX, at 44 degrees elevation and sinking.
0:22 Running copointing and dofpia on Veillet OB1
0:40 Veillet OB2 Slew & Execution, LBCR ~1.2\x94, LBCB ~1.4\x94
0:55 Veillet OB3 Slew & Execution, LBCR ~1.0\x94, LBCB ~1.3\x94
01:06 Veillet OB4 Slew & Execution, LBCR ~1.1\x94, LBCB ~1.4\x94
01:11 Veillet OB5 Slew & Execution, LBCR ~ 1.2\x94, LBCB ~ 1.5\x94
01:16 Veillet OB6 Slew & Execution, LBCR ~ 1.2\x94, LBCB ~ 1.2 \x94
01:20 Reconfiguring to MODS1 MODS2
01:40 Problems with SX shell, David restarting
01:51 Back in business, pointing checks, collimating
02:19 Slew to FF1725+3709, Aquisition
02:46 MODS2 FF1725+3709 science obs started, pointing adjustment made to MODS1 & check
02:49 MODS1 FF1725+3709 science obs started.
03:54 Slewing to standard, BD284211
03:55 SX primary panic
04:02 Active optics corrections removed to help with mirror recovery.
04:08 12 deg twilight.
04:14 Starting science
04:25 Finish obs, closing. Powering off LBCs
04:41 Starting MODS1 & 2 calibrations: Grating Arcs, Slitless Flats, & Biases