in Instrumentation Web
Search all webs
Results from Instrumentation web
retrieved at 14:39 (GMT)
The team: JHeidt (LSW): Lead, coordination everything WSeifert (LSW): Co lead, coordination everything SEsposito (Arcetri): Lead AO team GAgapito (Arcetri): A...
30 Jan 2015 - 07:08
Notes taken while troubleshooting Blue Filter Wheel 1 Error (IT 7294) Following are some notes which I have been collecting during the troubleshooting of IT 7294....
25 Jun 2018 - 16:55
UT 20161123 LUCI1 AO Nighttime Commissioning/FLAO tests Logs from JPower Observer LBTO: JPower, JHill Observer AO: DMiller Telescope Operator: DHuerta Mountain...
23 Nov 2016 - 15:50
UT 20170922 LUCI1/2 AO Nighttime Commissioning Observer LUCI Team: JHeidt, OKuhn (LBTO) Observer AO Team: DMiller, GTaylor (LBTO) ...
22 Sep 2017 - 12:07
UT 20170924 LUCI1/2 AO Nighttime Commissioning Observer LUCI Team: JHeidt, DThompson (LBTO) Observer AO Team: DMiller, GTaylor (LBTO) ...
24 Sep 2017 - 12:15
HEB temperatures slowly rising If the HEB temperatures are slowly rising, first confirm that the glycol flow and temperatures are ok. If so, then the thermo elect...
18 May 2015 - 16:56
Instrument Cooling System Troubleshooting There is a debug tool on the mountain for reading the glycol flow rates. The labels are not completely intuitive, but i...
14 Apr 2014 - 17:06
Inspection of MODS Masks Inspection of Cut Masks Masks should be inspected prior to installation in the instrument on the light table with the microscope or any ...
16 Nov 2016 - 18:46
LBC Lab Setup Instructions (Obsolete) Initial setup * It is assumed that the camera to be tested is ready (pumped and cold). * Set up LBC Lab Power Supply ...
28 Jul 2020 - 18:32
LBC Troubleshooting Please feel free to update this page with any LBC troubleshooting notes or links as appropriate. Thanks! * Troubleshooting notes from the ...
28 Sep 2022 - 20:31
Troubleshooting the LUCI1 NX Server The LUCI1 software runs on a SUN V880 Server, accessed via an NX Client. The machine name is lucifer.mountain.lbto.org or sun...
02 Jun 2014 - 21:29
MODS Mask Exchange Procedure MODS has a multi slit mask cassette that accepts up to 24 masks mounted in special frames. The masks are laser cut in thin spherical...
30 Nov 2021 - 19:48
PEPSI Important Links and Documents * Exposure time calculator * AIP PEPSI Users Manual Webpage * PEPSI Users Manual v1.0 is located in the Pepsi Documen...
23 Sep 2021 - 19:08
LBC Cryostat * Cryostat pumping and cooling procedure * Dismounting cryostats from telescope LBC Hub Focus Adjustment * Photo of position adjusting bolt...
26 May 2022 - 17:09
LUCIFER Reference Documentation Operating LUCI * Operating Manual * LUCI Cookbook link * LUCIFER Workstation Startup Procedure LUCI Installation and Re...
29 Sep 2020 - 21:17
LUCI1/2 Commissioning AO Com LUCI2 Diffraction limited Commissioning plan LUCI2 Diffraction limited Commissioning plan v2.4 as pdf Simones outline of LUCI2 FLAO...
12 Dec 2017 - 10:56
MODS: The Multi Object Double Spectrographs Overview MODS1 and MODS2 are UV to nearIR Multi Object Double Spectrgraphs that are mounted at the Direct Gregorian f...
28 May 2021 - 21:22
Troubleshooting Procedures for MODS team and LBTO staff These procedures are documented for use by MODS team and LBTO staff. They were separated from the Troubles...
08 May 2022 - 21:52
Number of topics:
Copyright © by the contributing authors. All material on this collaboration platform is the property of the contributing authors.
Ideas, requests, problems regarding Foswiki?