if (P11 and P12 reading are close to setpoint) then open AdSec cooling circuit valves and close bypass
if not: WARNING!! goto Ranfagni;
Switch circuit breakers on: they are located inside the Tower Laboratory. Switch on sequence: 110V first, then 380V
Switch ANTILEMP circuit breakers on, from web page http://18.104.22.168/ ; enable lines 2, then 3; log off, log on again and verify lines status
if (lines 2 and 3 are on (red spots on the web page)) then the remote circuit breakers are correctly switched on
if not: WARNING!! goto Ranfagni
Ping ADAM and wait for response.
End of switching on procedure
Starting AdSec software
ao_restart.sh, from terminal or pushing button on desktop: all active processes are killed, then restarted again;
press "Continue"; then press "System Processes";
verify whether all processes are running: ATTENTION HOUSEKEEPER led will remain yellow until AdSec is powered on;
Processes logs are stored into $ADOPT_LOG folder: such folder is /tmp on aospare
Adsec startup ????? (non ne' gia' partito?)
open AdSec Arbitrator GUI and press "PowerOn".
Open ILDCTRL log and verify the process completion
wait until HOUSEKEEPER led becomes green on System Processes window.
Open HOUSEKPR log and verify that HOUSEKEEPER begins logging values.
ATTENTION!: humidity sensors warm-up will result in a series of alarm and warning. Normal conditions will be restablished after warm-up (10 seconds...)
Press "LoadProgram" button on AdSec Arbitrator GUI
verify from IDLCTRL log if errors are occurred during communication test (Fastlink Alignment):
if (num errors > 0) then an automatic reset is processed and an error message is returned: press "LoadProgram" again;
else wait for the DSP program to be loaded
AO GUI: when launched remotely, exports on the remote machine the AO Software and the System Processes GUIs launched during startup. No AO process is executed on aospare. It is launched by typing startAO.py