Today (Feb 8th) AGW is reinstalled after the W-board shimming done in clean room.
Some problems with the telescope delay things, and we have the complete system (AGW+RR) at about 14:40 local time. In the meantime we have done a complete functional check of the AGW using the internal source and we found no issues. The internal pupil wobbling is confirmed to be about 4.5 pixels.
AGW mounted with Barrone installed.
Opened Flowerpot. Acting to the top screw of the pennina. Screwing-in, the gap on the top between pennina mount and the flowerpot
After 3/4 turn screwing-in: 2 pixel vignetting of the pupil
Back to zero
After 3/4 turn screwing-out: 3 pixel vgnetting of the pupil
Back to zero + 1/8 turn screwing-in. Pennina aligned. Closed Flowerpot.
Pupil wobble measurement