Summary on MODS checkout:
delta theta is close to what it historically has been, ~-0.2 deg, which suggests that the rotator zeropoint does not need to be changed. We did MOS acquisitions on 2 masks: StoneO (red and blue acq) and StoneA (red acq). We can use the old value in PCSInstrument.conf and the old transform for MODS1.
Stars are seen in boxes in the first thru-mask image, so pointing origin in PCSInstrument.conf file is OK.
BD+284211 observed in all modes. Feige110 observed in dual prism/grating and red prism/grating. This tested 5\x94 slit acquisition and got useful standard star data on a clear (and I think photometric) night.
Dice5 repeated 3 times on Stone O (though images looked elongated earlier, these look OK)
Long-slit alignment to be done
MODS1 looks OK (but has cataracts\x85 the debris on the field lens is really ugly. We confirm that the Y=620 position for MODS1 long-slit spectroscopy puts the star well below the gunk. I hope it can be cleaned.
4:30 MODS1 mask snap, followed by imaging flats. Only got u before quit so we could do test preset. U_sdss mods1b.20171030.0008-12
5:40 Running IMCS Lock test/confirmation imcsTTFs.pro for MODS1 @60deg rot 207. (This was run yesterday evening and b/rconfig.tab tables were updated with new values which were similar to the old except for red dual prism, where yesterday imcslock timed out on first try.).
IMCSlock achieved on first try for all modes. Mods1b.20171030.00013-18 mods1r.20171030.0008-13
6:05 Pointing and collimation
6:24 rm507-3 froze up during StoneO target acquisition. Olga was able to take over NX session to obs2 on rm507-5.
6:20 Stone O rotator center PA=0 active script send for collimation. Always doing instconfig to lock IMCS but guiding only at first PA.
Active PA0: mods1b 19 mods1r 14 (elongated stars on-axis; wfs shows some evidence of spherical)
Track PA180: mods1b 20 mods1r 15
Rotcen from 19&20 -> x0,y0= (1555.6,1565)
Rotcen from 15&16 -> x0,y0= (1550,1570)
Track PA90: mods1b 21 mods1r 16
Track PA270: mods1b 22 mods1r 17
21&22 -> x0,y0 = (1539,1541)
16&17 -> x0,y0 = (1536,1547) unsure why the 90/270 values do not agree with 0/180 ones, but I saw this before\x85 offline will have another look. (Dave solved this! Both observations need to be track mode presets. So the true rotator center are the ones above for 21&22 and 16&17).
6:47 resending active preset at 90 to recollimate
Active PA90: mods1b 23 mods1r 18
Track PA270: mods1b 24 mods1r 19
18&19 -> x0,y0 = (1569,1537) but the geomap residuals are ~1-2 pix in x and 0.5 pix in y\x85
|Blue Rotator Centers|
|20 & 21||-90||1533.3||1538.6|
|20 & 22||+90||1541.8||1535.1|
|21 & 22||+180||1539.5||1541.0|
|21 & 20||+90||1533.4||1538.5|
|22 & 20||-90||1541.8||1535.1|
|22 & 21||-180||1539.4||1541.0|
|spread||all||8.5 pix||5.9 pix|
|Red Rotator Centers|
|15 & 16||-90||1528.9||1544.5|
|15 & 17||+90||1539.3||1539.3|
|16 & 17||+180||1537.0||1547.1|
|Reverse order yields same results, to within 0.1-0.2 pixels|
|spread||all||10.1 pix||7.8 pix|
R 20, 21
modsAlign - used all boxes and stars except for the two under the guide probe:
Offsetpointing -0.1867 -0.776 1.361 detxy rel
Fixed problem with mask table (updateSlits needed to be run to update mask table on MODS1 IE server)
New acquisition on Stone O field: 23 & 24
modsAlign now with only selected boxes, to avoid the two vignetted by the guide probe: 23 & 24
Offsetpointing -0.1815 -0.940 1.074 detxy rel
Confirmatory image looks pretty good 25
All stars appear elongated --- is this thermal? Spherical? David is calling AO support.
Meanwhile, we are trying a blue acquisition on this field.
19:45 Blue acq only Acq images 27, 28 mods1b.
Offsetpointing -0.2153 -0.951 0.973 detxy rel
29 confirmatory thru-slit image, looks good.
19:56 Sending dice 5 PA=0 script on stoneO
Detxy absolute offsets 20\x94
DetXY Abs x4: (-20,20),(20,-20),(-20,-20),(20,20),(0,0) mods1b 0030-44 & mods1r 0026-40
COMM error (setting \x93partner\x94 timed out)
20:30 Going to BD+28. Nice and clear. Taking standard in all modes. Airmass 1.023 and setting modsAlign offset (-0.417, 10.616). Reran modsAlign on confirmation image 43 and offset (-0.465, -0.070), Seeing just jumped up. 1.35 on guider and 1.8 on DIMM.
BD28 Acquisition Mods1r 41-43
BD28 Dual Grating mods1b 45-47 mods1r 44-46
BD28 Red Grating mods1r 47-49
BD28 Blue Grating mods1b 48-50
BD28 Dual Prism mods1b 51-53 mods1r 50-52
BD28 Red Prism mods1r 53-55
BD28 Blue Prism mods1b 54-56
21:10 bouncing between 0.8 and 1.85\x94 on guider, highly variable.
21:29 StoneA mos acquisition to confirm rotator zeropoint, mosAlign. 056-057 slit and sky image. Offset -0.1897, -0.671 0.333 detxy rel
21:41 Sent preset for Feige110. Seeing ~1.25\x94, but varying on short timescales from 0.9 to 1.8
Offsets -1.139, 11.189. Reran confirmation modsAlign. Offsets 0.088, -0.022
Feige110 Acquisition Mods1r 59-60
Feige110 Dual Grating mods1b mods1r 61-62
Feige110 Red Grating mods1r 63-64
Feige110 Blue Grating mods1b
Feige110 Dual Prism mods1b 59-60 mods1r 67-68
Feige110 Red Prism mods1r 65-66
22:14 David is doing a pointing check
22:18 Sent preset to Al\x92s NSI target. Forgot to change to pseudo mono mode. David is doing that now for us.
22:33 Mods acquisition sent. Failed because no guide star in patrol field
22:44 LBC data being taken. Going to try to manually take exposures for mods
22:48 Cannot manually take spectra because we cannot send offsets without a preset. LBC data is more important at the moment. Unable to guide non-sidereally on this target with mods. Guide star just flies out of the FOV
23:07 Decided to just manually take unguided images 300sec in g and i mods1b 61-62, mods1r 72-73. LBC data being taken in z simultaneously guided.
23:25 Right Primary nearly in limit. Decided to change from pseudo monocular to monocular lbc, clear active optics.
23:47 Took garbage slave mods images. MODS was not tracking, rotator not rotating. Mods1b 63-64 and mods1r 74-75.
23:51 Preparing to handover telescope. LBC images continue
0:10 Done taking LBC exposures, Emanuele is all set up for MODS1. He had authorization issues with LBTB and mods (ssh keys for LBTB account), so Olga set him up on the LBTO account
0:12 David has completed the reauthorization
|mods1r.20171026.0002.fits||VFLAT + r_sdss Dual Imaging Flat||Imaging||Flat||VFLAT||r_sdss||3.0||2.0||25K|
|mods1r.20171026.0003.fits||VFLAT + i_sdss Dual Imaging Flat||Imaging||Flat||VFLAT||i_sdss||1.5||7.0||20K|
|mods1r.20171026.0004.fits||VFLAT + z_sdss Dual Imaging Flat||Imaging||Flat||VFLAT||z_sdss||1.5||3.0||20K|
|mods1r.20171026.0005.fits||QTH1 + ND1.5 + ND1.5 Spectral Flat||LS5x60x1.0||G670L||QTH6V QTH1||ND1.5||1.5||1.0||27K|
|mods1r.20171026.0006.fits||VFLAT + Clear Spectral Flat||LS5x60x1.0||G670L||VFLAT||Clear||10.0||1.5||23K|
|mods1r.20171026.0007.fits||VFLAT + Clear + ND1.5 Spectral Flat||Imaging||G670L||VFLAT||ND1.5||5.0||1.5||33K|
|mods1b.20171026.0002.fits||VFLAT + u_sdss Dual Imaging Flat||Imaging||Flat||VFLAT||u_sdss||10.0||50.0||23K|
|mods1b.20171026.0003.fits||VFLAT + g_sdss Dual Imaging Flat||Imaging||Flat||VFLAT||g_sdss||4.0||12.0||28 K|
|mods1b.20171026.0004.fits||QTH1 + ND1.5 + ND1.5 Spectral Flat||LS5x60x1.0||G400L||QTH6V QTH1||ND1.5||1.5||16.0||31K|
|mods1b.20171026.0005.fits||QTH1 + UG5 Spectral Flat||LS5x60x1.0||G400L||QTH6V QTH1||UG5||1.5||7.0||27K|
|mods1b.20171026.0006.fits||VFLAT + Clear + ND1.5 Spectral Flat||Imaging||G400L||VFLAT||Clear||5.0||5.0||37K|
|mods1b.20171026.0007.fits||VFLAT + UG5 Spectral Flat||Imaging||G400L||VFLAT||UG5||10.0||2.0||32K|
consider setting t=1 instead