Tucson: Jenny, Barry, John
Mountain: Kevin, David
Summary: Tested SX AO bin1, 2, 3, & 4 reconstructors. AGw transform verification not ready tonight. Performed Al\x92s 7 Iris D night target with the hopes that if ready tomorrow we can do the verification during the D night.
Night priorities outlined: https://docs.google.com/spreadsheets/d/1q3hQ04UESRPDJTzQn5SQ1EOXBOee_conayY7WJYgpRQ/edit#gid=0
Took some Dit5 Ndit 5 darks for Doug luci1.20170913.0032-0036 luci2.201709.0007-0011
18:33 Kevin Opening Up
Some thin clouds
18:45 test preset successful
18:48 Test Br Sky L10037 L20012
18:50 Br Gam low L10038-42 L20013-17 (5K)
18:55 H Test L10043 L2 0018
18:58 H High 11-12K L10044-48 L2 0019-23
Hlow (5-6K) L10049-53 L2 0024-28
19:04 z Test L10054 L2 29
19:06 z high L10055-0059 L2 0030-0034 Clouds moving in during these flats. No good
19:13 Going to zenith, rot at 341 to Home xstage. Authorize luci1 only
19:23 Pointing check star, focus & collimating. Taking a long time to collimate
19:57 Greg lost NoMachine session to mountain
19:59. John mentioned that no high order corrections seem to be going though
20:01 Seeing 0.9\x94 on guider, We are good to go!
20:03 SX LUCI AO test - FF1725+3709 AOref 8.2mag
FATAL: Timeout slopecompcntl, Greg noted not live but is now. Resent
20:17 Illegal command for state AO Aribitrator ready. Got stuck with optimize gain. Greg kicked it and seems happy now
20:18 resending FF1725+3709. DImm 0.86\x94
Greg hit done and it safe skipped right away so 0060 is probably no good
20:34 Greg tried to restart stuff but decided restarts of processes were needed.
\x93There is something decloaking overhead\x94 Clouds forming.
20:47 Greg took a hammer to everything. We resent preset, but pointing has drifted and we failed to reacquire guide star. David is doing a pointing check.
20:00 resending FF1725+3709 DIMM 0.88\x94
Greg has some issue with NCPA but can keep loop closed
21:12 images starting, dimm seeingg at 1.36
Aaaaand a cloud moved in after 1 exposure.
21:18 Switching to PEPSIPFU for checking DX M3 range of adjustment for wfs pupil alignment.21:31 preset to M5_1539 (DX AdSec ripped) DX IE=+2 CA=+30 M3 Abs Tip = -1000 (present setting) DX IE=+26 CA=+60 M3 Abs Tip = -1400 DX IE=+39 CA=+70 M3 Abs Tip = -1600 (negative limit of Tip for DX M3)
22:24 The clouds have cleared for the most part, except some lingering in the south..so Barry a target in the south for LUCIAO to resume. reauthorize
22:25 doing pointing check and collimation 0.76\x94 on the guider, 0.9\x94 on the DIMM
22:35 Sent script for 21h31 with offaxis 11.5 Rmag off axis AO ref. Looks like we are getting some clouds. Moving to the north
22:42 Sent script to M103
22:53 W unit failed so slewed to 90, rotator to park.
22:58 going to IRAS23365+3604 (a little higher and less distance to go if W unit fails again)
23:09 running! 0.72\x94 on guider 1.1\x94 on the dimm, bin 3 L1_0062-88
Noticed that with sky offset commands that things are moving before the pause ao completes
0080 is an example
23:51 Illegal command AOArbitrator. LoopFault, IIF Offset Failed
23:53 Resumed from where script failed L1 0090-0095 Seeing 0.73 on guider, 0.88 on Dimm
COMPLETED Bin 3 target
0:07 Slew to IC 5117 AOref 9th mag, Bin 2 (right on the border of 1,2) 153 modes
Observer gui stopped updating on obs2 and rm525-1 at the same time (~0:19). [observer gui
stopped updating previous night, too] Took luci snapshot on both. Exposures continued.
1:00 Failed. Continued but with edited script (6\x94 offsets instead) = failed, had to resend preset.
1:04 Observer panels froze on obs2 and rm525-1. Resent to IC 5117. Bin 1 with NCPA actually working. L1 123
Loop opened after during 0127 because of seeing bump
Closed and restarted (0.52\x94 guider, 0.79\x94 on DIMM)
1:52 Safe skip during offset. Last image 00145
1:54 resent preset, error: left secondary hexapod not in position
2:07 Resent preset Bin1 L1 0146, skipped to J where script died.
2:10 AO arbitrator process croaked. L1 0148 loop open. Moving on (seeing 0.64\x94 guider)
2:13 Going to 7 Iris, non sidereal ACE track preset. Preset failed, slopecompcntl timeout
2:17 resent to 7 Iris, DIMM 0.91\x94, Blasted with BrGam. Switching to H2
2:33 Sent to 7 Iris H2 setup, Bin1, 400 modes, NCPA on 2.51x20 L1 0154-0163
2:46 FF0234 AOref of 14.5 mag, testing bin4, 36 modes Seeing 0.85\x94 on DIMM,
Start L1 0164
0166 probe off axis guider moved just out of parol field (who cares, maybe save for testing software)
0171 also probe off axis guider moved just out of parol field (offset abs x -7 did it I think)
0.87\x94 Dimm, 0.61\x94 on guider
03:16 - 0.89\x94 DIMM 0.64\x94 guider - airmass 1.2
0180 also probe off axis guider moved just out of parol field
Last image 0181
3:22 7 Iris again with H2 setup L1 0182-0185 DIMM 0.81\x94 Seeing
Not sure if L1 0186 is ok Offset XY command interrupted
03:43 - Asteroid was on edge of patrol field - suggests pointing is off
03:45 - John H. manually moved to get the target closer - Greg running RunAO so we can continue script without resending preset
3:52 resumed 7 Isis from where we left off. L1 0187-192
4:15 Standard SJ9105 H & H2 L2 0193-202
Binary possibly, Surprise! On edge of correction so more observation needed.4:27 7 Iris, one last go. Jump in 203 L1 0203-212