UT 20130522 Technical Observing - MODS Collimation Issues
Observer: DMiller, JHill, OKuhn (Tucson) ASkemer, VBailey (LBTO)
Telescope Operator: GBechetti (LBTO)
AO Support: JCGuerra (Home)
Telescope Support: (LBTO)
Instrument Support: (LBTO)
03:00 Primary temperature gradients. See Z11, Z22 etc. WFSC looks like donut. Missing two left
03:30 Turn on high order corrections. Too much aberration so converged to poor image. Clear Primary Act Opt. Helps a bit. Clear primary and secondary. Collimates well
04:18 Turn off High order corrections. Leave current high order correction on to see stability. Stable. Getting down to 48 EL. Move higher
04:32 preset, BS9152, GS=1, TS=1. Primary mirror panic. Outer loop stalled message from PMC.
04:45 preset. guider image is awful! Active Optics corrects. Collimates
04:55 mods1b.20130522.00005/6.fits 10/30 second exposures. 10s image has structure. 30s has fairly round images.
05:00 preset, BS9152, GS=1, TS=5. image 7/8 30/10 second still elongation x = 6.7, y = 9.8 pixel fwhm.
05:20 setFocus left 5. (from 0.0) get 611 + 192 + 164 nm. MODS image #9 Seeing a worse, but more round images
05:30 setFocus left -5. (from 0.0) MODS 10 still elongated
05:50 preset BS9152, GS=8, TS=2, /YAXIS Trefoil converged value change by 800nm, change to 9-2 recon. Stable
06:35 preset BS9152, GS=8, TS=2, PA=160 get x and y axis. Stable.
More tests that show the new reconstructors do not work, even with no vignetting. Back to 20130118 reconstructor.
- 22 May 2013