AO Com Work: 26 October 2010 UT
Observer: DMiller, SEsposito, MXompero, LBusoni, APuglisi, FQuiros
Telescope Operator: DHuerta (LBTO)
Telescope Support: JUrban (LBTO)
Instrument Support: TShih (LBTO)
BP4 ACTIVE closed dome tests
02:20 David shut down all GUI's, Subsystems and network server. He the switched the build to BP4 via the lbtswitcher. He also logged in to tcs5 and used lbtswitcher to change to build BP4
02:30 David restarted all networks server (from TCS GUI), but we noticed MCS was still running and there were several GUI's still running. So he shutdown all network servers, we logged into all machine, kill MCS and all GUI's.
02:40 David started all network server
02:47 Doug set GCS into simulation mode by setting noAGW and noAZcam to true in lbt.conf
02:50 David stopped and started GCSR
02:55 preset, M5_0881, gs=0, MODE=TRACK Success
02:57 preset, M5_0881, gs=0, MODE=ACTIVE Failed. GCSR is not in simulation mode and took an image with AZcam
03:00 Checked lbt.conf. I had set GCSL to simulation. Now set GCSR to simulation. David restarted GCSR
03:02 preset, M5_0881, gs=0, MODE=ACTIVE Success. Find a star in acquisition image and guide and wfsc
BP4 ADAPTIVEACE closed dome tests
03:02 preset, M5_0881, gs=0, MODE=ADAPTIVEACE Failed. ccd47 in wrong state
03:02 preset, M5_0881, gs=0, MODE=ADAPTVEACE Success, including presetAO
03:10 Alfio reset some stuff in AO supervisor to allow closed loop with flowerpot fiber (W unit test source) Retroreflector is not mounted
03:15 preset, M5_0881, gs=0, MODE=ADAPTVEACE Success
03:15 Michele called
03:27 irc_runao, false Failed. IDL file error. IDL did not like Alfio's 0 gain filename
03:29 Alfio changed the name
03:30 irc_runao, false Failed. IDL file error. IDL still did not like Alfio's 0 gain filename
03:32 Alfio changed the name again
03:33 irc_runao, false Failed. IDL file error. IDL still did not like Alfio's 0 gain filename. Actually need to send a new preset so names of new files are read.
03:34 preset, M5_0881, gs=0, MODE=ADAPTVEACE Success
03:36 irc_runao, false Success AO closed loop on the test source. The status boxes on the IIF GUI are all correct
03:39 irc_pauseao Success. The AO system is now paused. The IIF GUI shows AO as Open(Red) This should be Paused(Yellow)...maybe
03:39 irc_resume Success. Loop closed and IIF status is correct (Closed).
03:41 iif_offset, 0.5, 0.0 Success. AO loop is paused, telescope slewed, AO loop is resumed. Is the GSC probe position also changed to reflect this offset? I could not tell on the GUI. Will try again later while watching the AGW probe position on the agw-control computer.
03:44 iif_preset, M5_0881, gs=0, MODE=ACTIVE Success This was to open the AO loop
Alfio says there is an offset without the pause and resume from IIF. Is this implemented?
04:49 iif_preset, M5_0881, gs=2, MODE=ACTIVE Success During closed dome test last week, off-axis guide stars caused a big slew and failure. May have been too far off-axis (outside the guide probe range, but GCS simulator did not care)
04:52 iif_preset, M5_0881, gs=2, MODE=ACTIVE, POSANGLE=180 Success. The guide probe can not move this position, but still successfull
05:10 Test WFSC "Power Off" button on AOS GUI. Successful
05:11 Test ADSEC "Power Off" button on AOS GUI. Successful
05:12 Test WFSC "Power On" button on AOS GUI. Successful. Took about 5 minutes
05:20 Test ADSEC "Power On" button on AOS GUI. Successful. Took about 1.5 minutes. Shell is against reference body.
05:23 Test ADSEC "Set" button on AOS GUI. Successful. Took about 30 seconds
05:30 David reactivated AO5. Took about 10 minutes from start of switch to ready for a preset. Quite Good!
05:40 Ready for a preset. Still closed due to high winds: 20 m/s
Humidity 25%, wind 16m/s
08:45 Open the chamber
08:51 Slew to a BS star and David acquires a star and set IE=-36 CA=37
09:00 pointto BS* in ACTIVE mode. Collimate to 500 nm rms
09:10 iif_preset, AO_04p00m08_2, gs=2, MODE=ACTIVE, /YAXIS
09:12 pointto M5_0162 ACQUIRE find star with IE=-75 CA=46
09:20 iif_preset, AO_04p00m08_2, gs=2, MODE=ACTIVE, /YAXIS
09:24 Collimated and AO team takes over.
A bunch of AO data on this star
12:15 iif_preset, AO_08p00m08_0, gs=4, MODE=ACTIVE Failed with no guide star
We then went an M5 star and set IE and C
iif_preset, AO_08p00m08_0, gs=4, MODE=ACTIVE Failed with no guide star again. Clear Active Optics and thin shell corrections on Secondary hexapod. preset again, but still nothing
We then went another M5 star and set IE and C
12: 45 iif_preset, AO_08p00m08_0, gs=4, MODE=ACTIVE Success
Collimated for a few minutes and then AO Team took over
==> Note, presetting to a new star should not take 30 minute. We need to clear Active Optics and Thin Shell corrections from M2 hexapod before slewing to a new star
==> Collimation is taking 10 iterations because the there is not lookup table being applied to the thin shell due to the gravitational induced astigmatism. We really need a lookup table. Can on be applied to the primary? We need Z6.
- 26 Oct 2010