|Installed in LgswUnitSx.
|back to MPE for servicing.
|entrance window cleaned. Camera housing opened in clean room for examination. Nut of water cooling inside the housing checked.
|Camera housing opened again for inside cleaning (glue debris found inside after picture examination). Normal check of camera operation.
|Impossible to start up the camera: temperatures between PEC and HK not consistent, see http://aowiki.arcetri.astro.it/ARGOS/Diary20140828
|While tightening the camera hoses a locking nut rotated and the camera Pressure started to rise. Leakage is suspected...