We received instructions from Julian by email on 11/05/2016:
To change the IP, you have to connect to the serial port of the controller (power supply rear side).
A USB adapter is somewhere in the cabinets on 3U, most likely close to the laser service notebook.
Baud rate is 115200.
After a restart of the controller, you have 3 seconds to send enter.
With the command "show" you can see the current configuration.
To change the ip, you send "ip=192.168.29.51;". Same syntax as shown, including the semicolon at the end.
Maybe you have also to adjust the gate etc.
When done, you leave the configuration by sending exit.
- 11 May 2016