- JZ fixes a fuse that was burnt because of the new piezo electronics required too much power.
- Checking for the poweron - poweroff procedures from arbitrator if a fuse fails the system could not be switched down...
- Bug fixes in the AARB (ask JB for details)
: switchon LgswSystem
SX and DX, we try to revive the APDs...
Both SX and DX systems fail to close the loop, the TT diverges.
We focus on SX: the APD3 fiber seems misconnected or broken, we go up to check:
We try to acquire a new dark, but now also ADP2 seems to not work properly: with flat illumination the counts are a factor 100 less than other channels.