: we unpack the PCD and the amplifier from 3L boxes, we bring it to 5. We also spend 1:30h in searching for the WfsCameraPump1
to evacuate the PnCCD
Jose applies a patch to LGSW housekeeping: now THORs on SX take about 15s to reconnect after booting. He also fixes Zabbix (several sensors where missing).
: start pumping PnCCD
. In the meanwhile we install the PCD and ampli on DX-LGSW (it has a different IP, Julian sent us an email about HowToChangePockelsCellAmplifierIp
We keep the pump on for about 2h, at the end the gauge reads 0.35 mbar. After 5min it reads 0.4mbar...
AOS v15.5 check w/ AARB
While pumping we check for the new AOS version.
: to login as tcs@tcs1 we must first login as telescope on an OBS machine.
We login as argos on FLAO on both DX and SX. The AARBs are on. On DX we have to revert to the default config file following instructions from Luca. After this all sides seem to be fine.
- On DX the AOS GUI says the LBTIWFS is connected
New LAT focus check
: we move to horizon, Gustavo must be trained to install and remove the AGW cubes.
While at horizon we also check for the LAT power supply indeed the basdard cannot connect to it. Power and ethernet to the MOXA seem fine. After a couple of checks Jose try to connect to it through the LATClient. He sends a CamPowerOn
command and the LAT basdard turns green.
- We must dig out why the basdard cannot connect to LAT at fresh start.
In the meanwhile Jose also fixed a problem with the SVN repository: Florian changed the SVN server, the repository had a new id for several hours, then he hacked it to get back the old id...
Sebastian and WG check for new LAT focus stage...
: LBC filling, it takes about 1h.
: PCT on SX-LGSW:
CL test of LGSW + PYR BIN2
: back to Zenith, we deploy the cal swing arm. Luca has restored the old AOS version.
We have problems in finding light because the tertiary selector mirror didn't home properly. Once fixed we get back light with hexapod in the March position:
| SX HEXA
The SX system was started from arbitrator, FLAO operated from argos account. TS service started properly.
We first close the loop with LGSW only, we center the FLAO stages on the on-axis source position.
| SX BOARD
- The TS seems to not work properly, ERROR addind temporary modes is displayed in the TS log. Restarting the process fixes this.
Then we close the loop with Pyr + LGSW. On axis source equivalent to mag 12 star. PyrWfs
set to Bin2 990 Hz. Truth sensing is integrating 990 frames (1s integration).
We change the on-axis source to mag 14. PyrWfs
set at Bin4 300Hz. TS integrates 300 frames (1s integration).
We switch to QC, keeping mag 14. PyrWfs
is set to Bin4 100Hz. TS integrates 500 frames equivalent to 5s integration.
- 11 May 2016