In the morning we have a residual Pie-Shape from the LBTI night.
recovers. Start of activities with cal. unit, we calibrate the FLAO with the on-axis source.
: FLAO REC wiht bin 2, TT only, KL_v16
We restore the Yellow reconstructor on ARGOS. FLAO slopes repositioned in the REC to be after # 1200.
We reverse the ARGOS REC to have a positive gain instead of negative (Tommaso reverses all the recs recorded so far).
: FLAO + ARGOS yellow reconstructor reversed
Yellow fails so we put in the Blue pupil.
: FLAO + ARGOS blue reconstructor reversed
The FLAO+ARGOS loop diverges so we reload just the ARGOS blue REC (still reversed). Also in this way the loop diverges.
while the ARGOS blue loop is divergin we get a Pie-Shape.